Models and types of flow meter that populated the market is quite varied where types can be adjusted to flow meter’s functions and flow meter’s applications in the field. These matters are worth noticing to be used as our reference when choosing types of flow meter that suits our needs.
Ultrasonic flow meter is a measurement tool with method of indirect measurement (inferential), because sensor does not measure the speed of the fluid in the pipe directly, instead, it measures different in time of transit (differential transit time), by measuring time of transit of the sound pulse with high-frequency (high-frequency sound pulses) which travels in the pipe with fluid flow. Time of transit or transit time is the time needed by the sound pulse to travels along the pipe in two directions, in the same direction as the flow and the opposite direction. The difference in time between the two equals the average speed of the fluid flow. Because of its measurement based on transit time, this method also called ultrasonic transit time flow meter.
Ultrasonic can be defined as many things. Since Portable Ultrasonic flow meter became popular, technicians think that flowmeter is a Portable Ultrasonic flowmeter. However, the ultrasonic flowmeter is a tool that is needed to measure flow velocity, volume, mass flow from liquids or gases with characteristics which can be moved around.
There are numerous companies that tried to develop this technology to be applied on steam, but so far the application of this tool is still uncommon and very limited. Ultrasonic flow meter so far has been made in various types and applications ranging from transit time or Doppler as base flow or liquid flow, low anemometer to describe the high cost of natural gas as a housing transfer application. The price for an ultrasonic flow meter can range from USD 25 to USD 150,000 depending on the application, and there are also handheld ultrasonic flow meters.
Principle of Operation
In this case, Doppler flow meter has Doppler shift which measures using continuous-wave in constant frequency or variations such as changes happening inside the Doppler shift is measured using vibrations.
In Ultrasonic flowmeter, transmission from a wave that passes the sensor indirectly connected with fluids is called non-contact type. There can also be misconceptions or simplification on transit time flowmeter.
Some information classified the new type as “phase shift” and compared to the older type which in starting meter uses “transit time“ and other use “Correlation transit time“.
Now a combination of the two transits is now made with ultrasound technology, this makes it possible to know exactly how many functions and calculations of the actual technique.
Ultrasonic Flowmeter type is this is quite popular especially for those who want the type of flow meter that can be moved around or in other words it is portable and it is also referred to as a Hand-held flow meter.
Its operations method is quite simple, just clamp it on to the pipe in which flow is going to be measured. For the output of the flow meter, while it can only be read on the display, there are also those equipped with printers so that the measurement results can be directly stored as hard copy, and there are also measurement results that can be stored on the memory card in the form of soft copies that can be read directly by the computer using a card reader.
2. Glass Tube Flow Meter
3. Thermal Mass Flow Meters
4. Coriolis Mass Flow Meter
Coriolis mass flow meter is a flow meter that directly measures the mass of fluids both liquid and gas. Sehingga Mass flow meter Coriolis cannot be affected by changes in temperature, pressure, viscosity, or even density. The basic method of Coriolis flowmeter is there is Coriolis force produced because the flow traveling inside the oscillating tube generates vibrations.
Force produced by the vibrations will be caught by a sensor which then will configure through Coriolis transmitter and produced linear flow. Coriolis flowmeter is highly accurate up to 0.1% and capable to operate in both high temperatures and pressure.
Coriolis mass flow meter in its application, will not be affected by changes in temperature because of the characteristic of Coriolis flowmeter the measures mass, so bubbles or air trapped inside the flow will not affect the measurement, because liquid mass measurement will ignore air.
5. Positive Displacement PD Meter
6. Turbine Flow Meter
Turbine Flowmeter basically uses the same principle as Woltmann rotating vane meter, where vane or turbine or impeller is placed inside the Flowmeter. This vane will rotate when fluid flows into the flow meter and make it very accurate when used to measure fluid with the slow flow but with really high pressure.
7. Electromagnetic Flowmeter
Electromagnetic Flowmeter is most widely used in water and waste and chemical flow measurement applications. Most of the applications of Electromagnetic flow meter are for industries such as the food, beverage, pharmaceutical, hospitality, and waste processing industries because they have to use a flowmeter that meets certain sanitation requirements.
8. Orifice Flow Meter
9. Laminar Mass Flowmeter
11.Vortex Flow Meters
|untuk info product dan harga silahkan hubungi : email@example.com|